KERRO KPE-103 electrode is made of PH glass electrode and a reference electrode combined composite electrode, which is a PH meter measuring sensor for measuring the concentration of an aqueous solution of H (PH value), it is widely used in various fields to be detected in the PH. The electrode you are buying is a precision test sensors elaborate analysis, after thorough testing, as the user carefully follows these instructions before operation and maintenance of products manufactured, will be able to obtain the best test results and extend the life of the electrode.
General purpose, standard size glass bulb pH sensing electrode
? Rugged, polycarbonate construction
? Glass pH sensing bulb surrounded by protective teeth
? Measuring Range: 0 to 14pH
? Connector: BNC, Probe: 1 Meter
KPE-103 can be used for all kinds of application of pH measurement and uses the universal BNC Connector with 1 meter probe.
|Model No||KPE-103 (KERRO)|
|Measuring Range||0 ~ 14pH|
|% theoretical slope||(PTS) =98.5% (25 ?)|
|Probe Length||1 meter|
|Resistance||=250MO (25 ?)|
|Alkali error||=15mV (25 ?)|
|Ambient Temp. (Operating)||0ºC~80ºC (32ºF~152ºF)|
|Relative humidity (Operating)||=85%|
Maintenance and precautions:
1. Test before removing the electrode protective sleeve (kit solution is 3MKCL, if there are crystals oozing is normal and does not affect the use of electrodes)
2. Observe whether all sensitive bulb filled with liquid inside, if there is a bubble, you should gently shake the electrode downward (like rejection thermometer) to remove air bubbles inside the sensitive bulb, otherwise it will affect test accuracy.